UPDATE 11/11/2014: The new station was calibrated and worked for about 24 hours before it failed again. The station will be down indefinitely until staff can get back on the roof to troubleshoot.
UPDATE 10/31/2014: The new station has been installed and is awaiting calibration.
UPDATE 9/2/2014: The city has purchased replacement parts and is waiting for them to be shipped and installed. NOTE: The anemometer (wind speed) and the rain collector bucket on the weather station on the roof of the Littleton Center were damaged in an early spring hail storm.